Agilent / HP / Keysight 8514B
8514B S-Parameter Test Set 45 MHz to 20 GHz, 3.5 mm
The Agilent 8514B s-parameter test set is available for the Agilent 8510C network analyzer for broadband coaxial measurements from 45 MHz to 50 GHz.
The Agilent 8514B test set has an architecture that develops a separate reference channel for each incident port. RF switching is done with a built-in electronic switch.
For active device measurements, the test set includes the ability to apply dc bias (external) to the test port center conductors. Also available are two 90 dB step attenuators, which allow control of the port 1 and port 2 signal levels.
45MHz to 20GHz
Separate reference channel for each incident port
Built-in electronic switch
Frequency Range: 0.045 to 20 GHz
Nominal Operating Power Level: 2 to -6 dBm
Connector Type: 3.5 mm, male
Impedance, DC Bias: 50 ohm nominal, 500 mA, 40 Vdc maximum
Attenuation Range (incident signal):0 to 90 dB, in 10 dB steps
Reference Power (dBm) at Port 1 (nominal): +2 -2 -6
Minimum Power (dBm) Measured at Port 2: -66 -95 -95
Receiver Dynamic Range (dB): 86 106 105
System Dynamic Range (dB): 68 93 89
Dynamic Range (for transmission measurements):
Frequency Range (GHz):.045 to 2 2 to 8 8 to 20
Maximum Power (dBm)Measured at Port 2: +20 +11 +10
RF Input Connector (rear panel):
Maximum Input Power: +16 dBm
Connector Type: 3.5 mm, female